USER’S GUIDE. Vaisala DEWCAP. ®. Precision SAW. Hygrometer. DM No part of this manual may be reproduced in any form or by any means, electronic. Manuals and User Guides for Pitney Bowes DM We have 2 Pitney Bowes DM manuals available for free PDF download: Operator’s Manual, Networking. View and Download Leica DM user manual online. DM Microscope pdf manual download.
|Published (Last):||14 January 2010|
|PDF File Size:||10.64 Mb|
|ePub File Size:||9.76 Mb|
|Price:||Free* [*Free Regsitration Required]|
Safety concept The individual modules of the Dj500 DM micros- You can combine individual system articles with copy series include an interactive CD-ROM with articles from external suppliers. Please read the all relevant user manuals several languages. Failure to comply can cause the following: Important Notes Description User Manual Accessories from third-party suppliers The Leica DM and DM B microscope meets This user manual includes important instruc- You can combine individual system articles with today’s mannual of the art of technology.
Neverthe- tions related to operating safety, maintenance articles from external suppliers. Protect contrary to the specifications in the user illumination to improve the visibility of objects.
LEICA DM USER MANUAL Pdf Download.
To parts may be used. Unplug the instrument before changing any fuses. These items are delivered in separate packages.
They should be used for safely storing and transporting the instrument should the need arise. Insert the dovetail into the stand support Type 1: Leica EZ viewing tube with integrated and tighten the set screw firmly without eyepieces.
Pitney Bowes DM500 Manuals
This precisely positions the view- ing tube onto the optical axis of the micro- scope, dm50 of the selected rotation. Leica EZ Viewing Tube — integrated eyepieces 1. To use the captive thumbscrew, completely 2. Position the Leica EZ viewing tube back in remove the set screw that comes with the place on the stand. Leica EZ viewing tube with integrated eyepieces.
Leica EZ Viewing Tube — integrated eyepieces cont. You can now safely mankal the Leica EZ viewing tube by loosening the captive thumbscrew, rotating the viewing tube, and tightening the captive thumbscrew again. The eyepieces are already integrated with the Leica EZ viewing tube and are pre-adjusted for corrected vision; Leica Standard Viewing Tube — separated eyepieces 1.
Insert the eyepieces into the eyetubes. Lock the eyepieces into the eyetubes by tightening the silver screws on the under- side of manua, eyetubes with a standard Cross type Phillips screwdriver not provided.
In this case, go to section room light. If you purchased your Manuual DM or DM B by components and not by standard configuration, please continue to section “Installing Objectives” on page No additional instal- lation or alignment is required.
As you rotate the nosepiece clockwise, Set the illumination control knob base.
Never use an adapter between the power 1. Turn the microscope on using the switch on the rotating ring aligns with the objective the bottom right-hand side of the micro- magnification in use. Place a specimen slide on the stage by slid- slide so a portion of the specimen is under ing the specimen slide manua, into the the objective in use. Slide grips hold the slide in place.
Fold or unfold the eyetubes to If you are using a Leica EZ viewing tube, which Eyepieces decrease or mankal the distance between already has the eyepieces integrated with This corrects for any vision differences between your right eye and left eye.
Place a drop of Leica Immersion Oil on the 5.
Raise the stage slowly using the coarse 1. Find the field of view on the slide that you specimen slide on the area of the specimen focus knob until the drop of oil on the spec- would like to inspect.
When finished using the oil immersion objective, be sure to clean the front of the objective, the specimen slide, and any other surfaces that the oil contacted using the instructions in section “Care of the Microscope” on page Now all you need to do is change objectives, set the condenser appropriately for the objective magnification in use, and enjoy the view!
Therefore, no lamp changing concave front lens of fairly short radius or is required for the life of the microscope. Also See for DM User manual – 46 pages. Page of 44 Go. Table of Contents Add to my manuals Add. Page 9 In order to prevent damage from vibrations, Unpacking These items are delivered in separate packages. Get Ready Get Ready! Eyeguards In this case, go to section room light.
Get Set Get Set! Using The Condenser 2. Page 31 This corrects for any vision differences between your right eye and left eye. Comments to this Manuals Your Name. Enter text from picture: By pressing ‘print’ button you will print only current page.
To print the manual completely, please, download it. Don’t show me this message again. Got it, continue to print. Don’t have an account?